"I synthesized some 18A, an 18-amino acid amphipathic peptide ... The HPLC purity analysis indicates that the peptide is now >95% pure. Pretty good if you ask me! ... I then connected two identical cartridges inline through the luer fittings. I used the exact same gradient, as measured in column volumes, the same flow rate (no change in linear velocity), and injected the same crude peptide volume for the new “double” cartridge purification and selected run."
Find out what she learned in the post Can I improve my peptide purification by increasing the column length? on The Peptide Synthesis Blog ».
Video Demonstration: Multi Racks for TurboVap®
Effective Purification of Peptides using Normal Phase Flash
Webinar Recording - Advances in Sample Preparation Using In-Well Hydrolysis for Urine Samples
Video Demonstration - the Software in Turbovap® Evaporators
A Holistic Approach to the Peptide Workflow in Drug Discovery